- Open Access
An SU-8-based microprobe with a nanostructured surface enhances neuronal cell attachment and growth
© The Author(s) 2017
- Received: 31 August 2017
- Accepted: 1 December 2017
- Published: 6 December 2017
Microprobes are used to repair neuronal injury by recording electrical signals from neuronal cells around the surface of the device. Following implantation into the brain, the immune response results in formation of scar tissue around the microprobe. However, neurons must be in close proximity to the microprobe to enable signal recording. A common reason for failure of microprobes is impaired signal recording due to scar tissue, which is not related to the microprobe itself. Therefore, the device–cell interface must be improved to increase the number of neurons in contact with the surface. In this study, we developed nanostructured SU-8 microprobes to support neuronal growth. Nanostructures of 200 nm diameter and depth were applied to the surface of microprobes, and the attachment and neurite outgrowth of PC12 cells on the microprobes were evaluated. Neuronal attachment and neurite outgrowth on the nanostructured microprobes were significantly greater than those on non-nanostructured microprobes. The enhanced neuronal attachment and neurite outgrowth on the nanostructured microprobes occurred in the absence of an adhesive coating, such as poly-l-lysine, and so may be useful for implantable devices for long-term use. Therefore, nanostructured microprobes can be implanted without adhesive coating, which can cause problems in vivo over the long term.
- Surface modification
- Nanostructured surface
- Nanosphere lithography
- Cell attachment
- Neurite outgrowth
Microprobes are used to repair or bypass brain injury by acquiring electrical signals from neuronal cells in contact with their surface. For reliable neural recording, a stable and sustainable interface between neuronal cells and the microprobes is required. A critical issue for microprobes is the difficulty in maintaining their recording performance for long periods , which is due to the biological response of the surrounding cells . Therefore, means of improving the lifetime and biocompatibility of implanted microprobes have been investigated .
Neural recording can be improved by applying a bioactive coating (such as of a protein or polymer) to the microprobe surface to enhance neuronal cell attachment. For example, Bellamkonda and colleagues used a thin coating of polyethyleneimine (PEI)–laminin (LN) to control scar tissue formation around implanted Si-based neural probes . The PEI–LN-coated silicon microelectrode arrays showed less gliosis at 4 weeks after implantation [4, 5]. However, attachment of neuronal cells to the microelectrode surface was not enhanced. A polyethylene glycol (PEG) coating containing L1 protein, which is a neural adhesion molecule, increased neurite growth and reduced glial adhesion eightfold compared to a laminin coating . A brain-derived neurotrophic factor (BDNF)—loaded hydrogel coating on planar microelectrode arrays (MEAs) affected neuron process growth and spontaneous electrical activity compared with control MEAs . However, the duration of release is dependent on the amount of protein loaded and the bioactive molecules in the coating; that is, it is limited. Therefore, further research is required to enhance our understanding of the microprobe–neuron interaction to prolong recording of neuronal signals.
Rough surfaces, especially nanostructured surfaces, maintain neuron viability more effectively than smooth surfaces, presumably because the former mimic the extracellular matrix [8, 9]. Various nanostructured surfaces—such as nanowire arrays [10, 11], nanopatterned poly-ε-caprolactone (PCL) , and a nanoporous gold (np-Au) substrate —increased neuronal growth compared with that on a flat surface. The morphology of np-Au selectively reduces astrocyte surface coverage while maintaining neuronal surface coverage, similar to unstructured planar gold surfaces . Ceramic-based microelectrodes using nanostructured porous silicon were proposed for use as a neural probe . Neurons preferred the nanostructured surface and extended a significantly greater number of neurites, while glial cells avoided the nanostructured surfaces. Subsequent in vivo studies showed that microelectrodes of nanostructured porous silicon could be used to record signals from single neurons ; at 1 week, glial activation was reduced and a greater number of neurons were adjacent to the porous silicon surfaces than to the smooth silicon surfaces. Microprobes made of soft materials as an alternative to silicon (Si) and ceramic-based neural probes have been developed and can suppress the inflammatory response after implantation. Although neuronal cells prefer a nanostructured surface, the majority of flexible microprobes have a flat surface. In our previous work, a nanostructured SU-8 substrate exhibited similar neuronal outgrowth to a flat SU-8 surface coated with poly-l-lysine (PLL), which facilitates adhesion by rat pheochromocytoma (PC12) cells [17, 18]. These findings motivated us to develop SU-8-based microprobes with nanostructures to support neuronal cells on their surface over a long period of time.
Here we present a method for the preparation of nanostructured SU-8 microprobes to improve neuronal binding and growth. We used SU-8 because of its biocompatibility and suitability for neural interface implants [19, 20]. We can apply the proposed procedure to the preparation of planar microprobes, as discussed in our previous work . Therefore, the microprobes were prepared by conventional photolithography and the nanostructures on their surface were then realized by nanospheric lithography (NSL) [21, 22]. In addition, a nanostructured surface can be patterned as desired simply by covering undesired areas with a thin layer of metal. Scanning Electron Microscopy (SEM) and Atomic Force Microscopy (AFM) were used to characterize the surface nanostructures. The adhesion and outgrowth of PC12 cells was improved on the nanostructured SU-8 microprobes. Our results suggest that nanostructured SU-8 microprobes provide a flexible neural interface that enhances their long-term performance and stability.
Fabrication of nanostructures on the surface of SU-8 microprobes
To fabricate nanostructures around the electrodes, a second SU-8 layer of identical thickness to the first was added using the same procedure (Fig. 2c). For the selective generation of nanostructures around the Au electrodes, a Cr layer was deposited and patterned by a lift-off process using a positive photoresist (AZ 5214) (Fig. 2d). For nanosphere lithography, the substrate with SU-8 microprobes was cut into 18 × 18 mm pieces, and coated with a monolayer of 0.3-µm-diameter polystyrene nanoparticles (PS NP suspension, 5.0% coefficient of variation; Spherotech, Lake Forest, IL, USA) by transferring the PS monolayer to the substrate by the water–air interface . The size of the PS nanobeads was reduced by oxygen plasma reactive-ion etching (RIE) (FabStar; Top Technology Ltd., South Korea) for 1 min with an O2 flow rate of 30 sccm and power of 50 W. A second 30-nm-thick layer of Cr was deposited in the spaces among the PS nanobeads using an evaporator (Fig. 2e). The PS nanobeads were dissolved in toluene at 40 °C for 30 min with sonication, and nanostructures were generated by RIE with an O2 flow rate of 30 sccm and power of 40 W for 4 min. The Cr layer on SU-8 functions as an etch mask for the fabrication of nanostructures. Finally, the residual Cr layer (Fig. 2d, e) was removed using CR-7 etchant for 1 min and the substrate was cleaned in deionized water and disinfected in 100% ethanol (Merck, Darmstadt, Germany) for 24 h in the dark. The substrate was washed three times in sterile distilled water and dried in a laminar flow hood.
Characterization of the nanostructured surface
A field-emission scanning electron microscope (FE-SEM) (S-4800; Hitachi, Tokyo, Japan) was used to observe the surface morphology of the nanostructured microprobes. An AFM (NX10; Park System Corp., South Korea) in non-contact mode in air at room temperature was used to measure the surface roughness and topography of the nanostructured surface.
PC12 cell culture
PC12 cells (CRL-1721; ATCC, Manassas, VA, USA) are extensively used in neurobiological and neurochemical studies [23, 24]. In this study, PC12 cells were used to investigate the responses of neurons to the nanostructured and non-nanostructured microprobes in vitro. PC12 cells were maintained in Roswell Park Memorial Institute (RPMI) 1640 medium with l-glutamine (Gibco, Grand Island, NY, USA) supplemented with 10% horse serum (Sigma-Aldrich, St. Louis, MO, USA), 5% fetal bovine serum (FBS; Hyclone, Logan, UT, USA, and Thermo Fisher Scientific, Rockford, IL, USA) and 1% antibiotic–antimycotic solution (penicillin 10,000 units/mL, streptomycin 10,000 μg/mL, Fungizone (amphotericin B) 25 μg/mL; Gibco Life Technologies, Grand Island, NY, USA) at 37 °C in 5% CO2 and subcultured at least twice weekly. For differentiation, PC12 cells were seeded onto plates at 2.5 × 105 cells/mL and treated with 50 ng/mL nerve growth factor (NGF; BD Biosciences, Bedford, MA, USA) for 7 days. NGF-supplemented medium was changed every 2 days.
Neurites longer than 25 μm (i.e., the average cell body diameter) were enumerated. Three areas were randomly selected on each substrate by scanning from top to bottom under a microscope. Experiments were conducted in triplicate. Comparisons were carried out using Student’s t test. Results are presented as means ± standard errors, and a value of p < 0.001 was considered to indicate statistical significance.
Fabrication of SU-8-based microprobes
Characterization of surface nanostructures
In vitro culture of PC12 cells on nanostructured SU-8 microprobes
To compare neuronal development between nanostructured and non-nanostructured microprobes, the number of neurites extended per PC12 cell was determined in phase-contrast optical images. Figure 5d shows the number of neurites per differentiated PC12 cell on non–nanostructured and nanostructured microprobes for 5 days after NGF treatment. Differentiation of PC12 cells was significantly greater on nanostructured than on non-nanostructured microprobes (p < 0.001). In contrast, few PC12 cells were attached to non-nanostructured microprobes, and those that did attach were in close proximity to each other (Fig. 5a). Because PC12 cells differentiate as they attach to the surface, similar to our previous study, they showed low neurite outgrowth over time at 5 days after NGF treatment . It is interesting to note that the length of PC12 neurite extension on nanostructured microprobes is similar to that on rat tail collagen type I coated culture dishes . As shown in Additional file 1: Figure S2, PC12 cells extended to the longest neurite length for about 150 μm 10 days after NGF treatment. Surface attachment and growth by PC12 cells typically require an adhesive coating (e.g., PLL), but nanostructured microprobes enhance cell adhesion and differentiation without the requirement for a surface coating. Nanostructures are permanent; in contrast, the effect on cell adhesion of a coating of adhesion molecules is limited by the inevitable degradation over time . Therefore, nanostructured microprobes support neuronal growth for longer periods than those with conventional surface coatings.
Further studies should determine whether nanostructured microprobes support neuronal growth over the long term in comparison with microprobes coated with cell-adhesion molecules. After microprobe implantation, the foreign-body reaction and inflammatory response should be evaluated to validate the utility of nanostructured microprobes in vivo. Nanostructured microprobes show promise for implantable brain-machine interfaces that can remain in situ for long periods.
In this study, nanostructured SU-8 microprobes were developed to enhance neuronal growth with the aim of facilitating long-term recording of neuronal signals. SU-8 microprobes were produced by conventional photolithography and nanostructures of ~ 200 nm diameter and depth were generated on their surfaces. Neuronal development on nanostructured and non-nanostructured microprobes was characterized using PC12 neuron-like cells. PC12 cells on nanostructured microprobes showed significantly increased attachment and neurite outgrowth compared with those on non-nanostructured microprobes. This improved neuronal development did not require an adhesive coating (e.g., PLL) and so the nanostructured microprobes show promise for use in implantable devices that support neurons, for example, implantable brain-machine interfaces.
EK carried out the experiment and drafted the manuscript. EK, JK and HC conceived of the study. All authors read and approved the final manuscript.
We thank the Center for Core Research Facilities of DGIST for technical support. The authors are also grateful to Sangwon Kim and Jongmoon Jang (DGIST) for their help with the SEM inspection.
The authors declare that they have no competing interests.
Ethics approval and consent to participate
Funding for this research was provided by the Global Research Laboratory from the National Research Foundation of Korea (NRF), funded by the Ministry of Science and ICT (No. NRF 2017K1A1A2013237), Korea Evaluation Institute of Industrial Technology (KEIT) funded by the Ministry of Trade, Industry & Energy (No. 10052980), and by DGIST MIREBraiN Project.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Abbott A (2006) Neuroprosthetics: in search of the sixth sense. Nature 442(7099):125–127View ArticleGoogle Scholar
- Moxon K, Hallman S, Sundarakrishnan A, Wheatley M, Nissanov J, Barbee K (2009) Long-term recordings of multiple, single-neurons for clinical applications: the emerging role of the bioactive microelectrode. Materials 2(4):1762View ArticleGoogle Scholar
- Polikov VS, Tresco PA, Reichert WM (2005) Response of brain tissue to chronically implanted neural electrodes. J Neurosci Methods 148(1):1–18View ArticleGoogle Scholar
- He W, McConnell GC, Bellamkonda RV (2006) Nanoscale laminin coating modulates cortical scarring response around implanted silicon microelectrode arrays. J Neural Eng 3(4):316View ArticleGoogle Scholar
- He W, Bellamkonda RV (2005) Nanoscale neuro-integrative coatings for neural implants. Biomaterials 26(16):2983–2990View ArticleGoogle Scholar
- Azemi E, Stauffer WR, Gostock MS, Lagenaur CF, Cui XT (2008) Surface immobilization of neural adhesion molecule l1 for improving the biocompatibility of chronic neural probes: in vitro characterization. Acta Biomater 4(5):1208–1217View ArticleGoogle Scholar
- Jun SB, Hynd MR, Dowell-Mesfin NM, Al-Kofahi Y, Roysam B, Shain W, Kim SJ (2008) Modulation of cultured neural networks using neurotrophin release from hydrogel-coated microelectrode arrays. J Neural Eng 5(2):203View ArticleGoogle Scholar
- Zhang L, Webster TJ (2009) Nanotechnology and nanomaterials: promises for improved tissue regeneration. Nano Today 4(1):66–80View ArticleGoogle Scholar
- Park J, Moon D, Kim J, Lee J (2016) Cell adhesion and growth on the anodized aluminum oxide membrane. J Biomed Nanotechnol 12(3):575–580View ArticleGoogle Scholar
- Piret G, Perez M-T, Prinz CN (2013) Neurite outgrowth and synaptophysin expression of postnatal cns neurons on gap nanowire arrays in long-term retinal cell culture. Biomaterials 34(4):875–887View ArticleGoogle Scholar
- Robinson JT, Jorgolli M, Shalek AK, Yoon M-H, Gertner RS, Park H (2012) Vertical nanowire electrode arrays as a scalable platform for intracellular interfacing to neuronal circuits. Nat Nanotechnol 7(3):180–184View ArticleGoogle Scholar
- Cesca F, Limongi T, Accardo A, Rocchi A, Orlando M, Shalabaeva V, Di Fabrizio E, Benfenati F (2014) Fabrication of biocompatible free-standing nanopatterned films for primary neuronal cultures. RSC Adv 4(86):45696–45702View ArticleGoogle Scholar
- Chapman CA, Wang L, Chen H, Garrison J, Lein PJ, Seker E (2017) Nanoporous gold biointerfaces: modifying nanostructure to control neural cell coverage and enhance electrophysiological recording performance. Adv Funct Mater 27(3):1604631View ArticleGoogle Scholar
- Chapman CA, Chen H, Stamou M, Biener J, Biener MM, Lein PJ, Seker E (2015) Nanoporous gold as a neural interface coating: effects of topography, surface chemistry, and feature size. ACS Appl Mater Interfaces 7(13):7093–7100View ArticleGoogle Scholar
- Moxon KA, Kalkhoran NM, Markert M, Sambito MA, McKenzie J, Webster JT (2004) Nanostructured surface modification of ceramic-based microelectrodes to enhance biocompatibility for a direct brain–machine interface. IEEE Trans Biomed Eng 51(6):881–889View ArticleGoogle Scholar
- Moxon K, Hallman S, Aslani A, Kalkhoran N, Lelkes P (2007) Bioactive properties of nanostructured porous silicon for enhancing electrode to neuron interfaces. J Biomater Sci Polym Ed 18(10):1263–1281View ArticleGoogle Scholar
- Kim E, Yoo S-J, Moon C, Nelson BJ, Choi H (2015) Su-8-based nanoporous substrate for migration of neuronal cells. Microelectron Eng 141:173–177View ArticleGoogle Scholar
- Kim E, Yoo S-J, Kim E, Kwon T-H, Zhang L, Moon C, Choi H (2016) Nano-patterned su-8 surface using nanosphere-lithography for enhanced neuronal cell growth. Nanotechnology 27(17):175303View ArticleGoogle Scholar
- Nemani KV, Moodie KL, Brennick JB, Su A, Gimi B (2013) In vitro and in vivo evaluation of SU-8 biocompatibility. Mater Sci Eng 33(7):4453–4459View ArticleGoogle Scholar
- Cho S-H, Lu HM, Cauller L, Romero-Ortega MI, Lee J-B, Hughes GA (2008) Biocompatible SU-8-based microprobes for recording neural spike signals from regenerated peripheral nerve fibers. IEEE Sens J 8(11):1830–1836View ArticleGoogle Scholar
- Jeong G, Park J, Lee K, Jang J, Lee C, Kang H, Yang C, Suh S (2010) Fabrication of low-cost mold and nanoimprint lithography using polystyrene nanosphere. Microelectron Eng 87(1):51–55View ArticleGoogle Scholar
- Ho C-C, Chen P-Y, Lin K-H, Juan W-T, Lee W-L (2011) Fabrication of monolayer of polymer/nanospheres hybrid at a water–air interface. ACS Appl Mater Interfaces 3(2):204–208View ArticleGoogle Scholar
- Fujita K, Guroff G, Yavin E, Goping G, Orenberg R, Lazarovici P (1990) Preparation of affinity-purified, biotinylated tetanus toxin, and characterization and localization of cell surface binding sites on nerve growth factor-treated pc12 cells. Neurochem Res 15(4):373–383View ArticleGoogle Scholar
- Greene LA, Tischler AS (1976) Establishment of a noradrenergic clonal line of rat adrenal pheochromocytoma cells which respond to nerve growth factor. Proc Natl Acad Sci 73(7):2424–2428View ArticleGoogle Scholar
- Offenhäusser A, Sprössler C, Matsuzawa M, Knoll W (1997) Field-effect transistor array for monitoring electrical activity from mammalian neurons in culture. Biosens Bioelectron 12(8):819–826View ArticleGoogle Scholar
- Reier PJ, Perlow MJ, Guth L (1983) Development of embryonic spinal cord transplants in the rat. Dev Brain Res 10(2):201–219View ArticleGoogle Scholar
- Mattila PK, Lappalainen P (2008) Filopodia: molecular architecture and cellular functions. Nat Rev Mol Cell Biol 9(6):446–454View ArticleGoogle Scholar
- Attiah DG, Kopher RA, Desai TA (2003) Characterization of PC12 cell proliferation and differentiation-stimulated by ECM adhesion proteins and neurotrophic factors. J Mater Sci 14(11):1005–1009Google Scholar
- Mingyu C, Kai G, Jiamou L, Yandao G, Nanming Z, Xiufang Z (2004) Surface modification and characterization of chitosan film blended with poly-l-lysine. J Biomater Appl 19(1):59–75View ArticleGoogle Scholar