 Letter
 Open Access
 Published:
Analysis of liquidtype proof mass under oscillating conditions
Micro and Nano Systems Letters volume 8, Article number: 12 (2020)
Abstract
In this study, the spring constant of an accelerometer with a liquidtype proof mass was analyzed. Unlike a general solidtype microelectromechanical system accelerometer, the Laplace pressure is considered a restoring force in the analyzed accelerometer. Using a base excitation mathematical model, the sensor output could be estimated for a specific spring constant. Although the estimated sensor output data fit well with the experimental results, the spring constant of the device could also be determined dynamically (for oscillations below 5 Hz). Moreover, the damping constants could be inferred depending on whether sandblasting treatment was performed. Finally, the effects of the oscillation, surface condition, and volume of liquid metal droplets on the spring constant were analyzed.
Introduction
Accelerometers generally detect acceleration by the movement of the proof mass in the device [1]. Various sensing mechanisms exist for detecting the location of a proof mass (e.g., capacitive [2,3,4,5,6], piezoresistive [7, 8], resonant [9, 10], and optical [11, 12]). Many devices are fabricated using microelectromechanical system (MEMS) processes. Such accelerometers are composed of solid springlike structures and can suffer from mechanical fatigue with prolonged usage [13, 14]. To improve the fatigue characteristics of solidtype MEMS accelerometers, accelerometers utilizing the properties of liquid metals (LMs) (e.g., high conductivity, density, and surface tension) have been developed [15,16,17].
In our recent study, a capacitivetype twoaxis accelerometer using an LM was developed [17]. Because the capacitive sensing mechanism involved a dielectric structure on the electrodes, surface modification was performed on the electrode layer. In terms of the mobility of LMs in a channel, surface modifications can effectively improve the performance of the device. However, to design the device for this purpose, the spring constant should be analyzed and predicted for various situations.
In this study, the spring constant of an accelerometer using LM was analyzed. Using a base excitation mathematical model, the sensor output could be estimated for a specific spring constant. The spring constant was determined by dynamically matching the estimated values from the base excitation model and the experimental results. The damping constant was also determined using this method. Finally, the effects of the oscillation, surface condition, and volume of LM droplets on the spring constant were analyzed.
Device configuration and sensing principle
The device comprised a sensing part, an aligner, a shielding electrode (copper tape), and acrylic cases. The sensing part included a sensing printed circuit board (PCB), pressuresensitive adhesive (PSA) film, LM, and a PMMA guiding channel. The fabrication process of the device is shown in Fig. 1a. First, a PSA film used as a dielectric layer was attached to a sensing PCB board. Subsequently, 8 μL of LM droplet was deposited on the PMMA guiding channel and then covered with the PCB board. Next, the channel and PCB board were fixed using the aligner. The surfaces of the PSA film and PMMA channel were applied with and without sandblasting treatment to compare the mobility of the LM droplet in the device. Sandblasting conditions of 0.3 MPa for 10 s and 0.1 MPa for 3 s were applied to the PMMA and PSA, respectively. The distance between the specimen and the nozzle of the sandblaster was approximately 7 cm. Because capacitancetype sensors are easily affected by parasitic capacitance, patterned copper tape was placed on the aligner and connected to a ground for shielding. Finally, the acrylic cases were covered on both sides and fastened with M3 bolts. Using a capacitancetodigital converter chip (AD7147, Analog Devices), each sensing cell measured ~ 20 pF in capacitance at ~ 330 Hz. The output of the counter varied linearly from 0 to 65,536 (16 bits) as the capacitance changed from approximately 4 to 25 pF.
The position of the LM was measured based on the floating electrode concept. The configuration of the sensing electrodes is shown in Fig. 1b; one set of the sensing part was paired with two cells (i.e., sensing and ground). Specifically, if the bottom cells are represented by G and S_{3}, Cap._{1} is the capacitance generated between the LM and G, and Cap._{2} is the capacitance generated between the LM and S_{3}. Cap._{3} represents the capacitance between the two bottom electrodes (i.e., G and S_{3}), which is constant. As the area of the LM covering the two cells widened, Cap._{1} and Cap._{2} increased. Therefore, the overall capacitance increased as well. As shown in Fig. 1d, two sets of electrodes were arranged in the x and yaxis directions, separately. The acceleration of each axis was measured by the difference in capacitance at both sets of electrodes (e.g., C_{x}= C_{xR}− C_{xL}; C_{y}= C_{yU}− C_{yD}).
When acceleration was applied, the difference in the radius of curvature across the LM droplets increased. Subsequently, the LM generated a restoring force to the center, which functioned as a spring [17]. The mobility characteristics of the LM improved owing to the surface modification on the PMMA and PSA films. This phenomenon was analyzed in terms of the damping effect (Fig. 2a).
Detailed information about the device developed in this study (e.g., characterization data and comparisons with other types of accelerometers) is provided in [17].
Determination of spring constant
Spring constant and characteristic area during oscillation
In a dynamic environment where an oscillation occurs, defining the acceleration at a specific location is difficult because the shape of the LM is not saturated. However, a sandblastingtreated device demonstrates an interesting phenomenon, wherein the LM vibrates at a natural vibration.
The setup of the oscillating device is shown in Fig. 2b. The device was attached to the end of the linear stage (PS012380R, LinMot). To observe the behavior of an oscillating LM droplet, the copper tape for shielding was removed, and the LM vibration was recorded as a video (Additional file 1: Movie S1). An analysis of the experimental data revealed that the vibration frequency was approximately 13 Hz (8 μL of LM) (Fig. 2c). To obtain the vibration frequency, the sensor output was measured using a highpass filter to remove lowfrequency signals. Based on the vibration frequency, the characteristic area in a dynamic situation can be calculated. In the case of the fabricated device, the restoring force of the proof mass is the Laplace pressure. Therefore, to obtain the spring constant, the pressure must be converted to force by multiplying it by the characteristic area. Assuming that the damping effect almost converged to zero by sandblasting treatment, the natural frequency in the springmass system is expressed as follows:
where w_{n}, k, and m represent the natural frequency, spring constant, and mass of the proof mass, respectively.
When w_{n} was approximately 13 Hz, k was 0.018 N/m for 0.10810^{−3} kg of m. Therefore, the characteristic area in the dynamic situation was approximately 0.01 mm^{2}, such that k was 0.018 N/m. Fitting can be performed by slightly adjusting the spring constant based on the spring constant for various situations (e.g., 5 Hz oscillation, high surface roughness, and a large volume of LM).
Base excitation model
To determine the spring and damping constants, the actual experimental result and the mathematical result obtained from the base excitation model were compared. The Laplace pressure generated from the LM droplet was described as springs. The detailed calculation of the Laplace pressure based on the location of the LM droplet is provided in [17]. Moreover, the increased mobility of the LM droplet in the sandblastingtreated device can be considered as a reduction in the damping effect. Furthermore, the base excitation mathematical model was used for validation (Fig. 2d). For specific spring and damping constants, the sensor output can be estimated using the mathematical model. When matching the estimated values calculated by the base excitation model and the experimental results in dynamic situations, the spring and damping constants can be determined. The base excitation model equation is shown in (2).
where m_{LM}, k_{Laplace}, c, x, and y represent the mass of the LM droplet, spring constant induced by the Laplace pressure difference, damping coefficient, absolute position of the LM droplet, and absolute position of the base, respectively. The absolute position of the base with time can be expressed as (3).
where F_{0}, w, and t represent the motion amplitude, frequency of base, and time, respectively.
The result of the base excitation model is defined as x(t)y(t), which is the relative position of the LM droplet at the base. The experimental setup (Fig. 2b) was established such that the reciprocating motion was exactly y(t). The signals of the device were observed while the 0.03 m of stroke was reciprocated by two frequencies (i.e., 2 and 5 Hz). The variables of the base excitation model were changed until the tendencies of the experimental result and the base excitation model were similar.
To verify the effect of the LM droplet size, 8 and 12 μL LM droplets were used for each test. F_{0} was determined as 0.03 m in stroke length, and the LM droplet remained at the edge of the channel for a moment by inertia when acceleration was applied. At this moment, the distance from the center of the channel to the center of the LM droplet was defined as the initial position. In addition, because the initial velocity differed according to the oscillating frequency, the velocity of the LM droplet was estimated according to the frequency ratio (i.e., 0.02 m/s at 2 Hz of w, and 0.05 m/s at 5 Hz of w). The damping constant was determined by matching the experimental data with the mathematical model calculation results. The parameters of the base excitation model were determined and are listed in Table 1.
As shown in Fig. 2e, for the oscillation of an 8 μL droplet at 2 Hz, the results from the base excitation model were similar to the experimental results (Fig. 2c). A comparison of the results from the experiment and the base excitation model is shown in Fig. 3. Most of the results indicated similar tendencies; the vibration of the LM droplet reduced, and the amplitude reduced when sandblasting was not performed (the damping effect). When the sandblastingtreated device was oscillating at 5 Hz, the LM’s vibration was not suppressed as the damping effect was absent. Therefore, the vibration of the LM droplet was canceled but superimposed on the movement of the LM droplet. Consequently, the peak exhibited a random motion. A similar graph can be obtained from the mathematical model results.
Reduction of spring constant
In our previous study [17], the characteristic area in a static situation was determined based on a mathematical model and a tilting experiment. The results showed that the characteristic area in a static situation with an 8 μL droplet was approximately 1 mm^{2}. As can be inferred from “Spring constant and characteristic area during oscillation” section, the characteristic area (~ 0.1 mm^{2}) in the dynamic situation was smaller than that in the static situation. The reduction in the characteristic area eventually resulted in a decrease in the spring constant. In this section, the variation in the spring constant is analyzed for various cases (i.e., oscillation, surface roughness, and volume of LM droplet).
Oscillation/dynamic situation
The characteristic area of an oscillation is less than that of a static situation (i.e., 1.0 and 0.1 mm^{2} for static and dynamic situations, respectively) because the time for stabilization is insufficient. For comparison, the behavior of the LM in a static situation is shown in Fig. 4a. As the LM droplet traveled, it instantaneously stretched in the direction of movement, and the curvature differences (i.e., R_{1} and R_{2}) at both its ends decreased (Fig. 4b). Consequently, the Laplace pressure difference decreased, and one might predict that the spring constant would decrease in a dynamic environment.
Surface roughness
The spring constant of a device with sandblasting treatment is larger than that without sandblasting treatment. The sandblasting treatment increased the surface roughness. In the sandblastingtreated device, the moving resistance of the LM droplet decreased on the upper and lower walls owing to the high surface roughness. On a surface without sandblasting treatment, the LM droplet exhibited better adhesion to the surface, and the LM stretched as it moved within the channel. Therefore, the curvature differences across the LM droplet on the surface without sandblasting treatment were smaller than those with sandblasting treatment, causing the spring constant to decrease (Fig. 4c). As shown in Fig. 3, when the device without sandblasting treatment was oscillating at 5 Hz, although the damping constant increased, the output was larger, which supported the prediction that the device without sandblasting treatment reduced the spring constant.
Effect of LM droplet volume
As shown in Table 1, the spring constant increases slightly with LM volume. The same results were obtained from the mathematical model in [17] (Fig. 5a). As the LM volume increased, the curvature difference between both ends increased as the LM moved. Although the spring constant increased, the sensitivity of the device increased, as shown in the experimental result (Fig. 5b). Based on the results, it can be predicted that the effect of mass is more dominant than the increment in the spring constant. In Fig. 5a, R_{3} represents the horizontal curvature radius of the LM (radius of LM from the top view). Therefore, an increment in R_{3} means an increment in LM volume.
Previous results indicated that an increment in volume improved the sensitivity. However, the volume should be optimized depending on the velocity range. If the LM volume is increased excessively, then the unstable behavior of the LM will increase in dynamic conditions. The instability of the LM can be explained by the Weber number (We), which defines the ratio of the kinetic energy to the surface energy, as in (4).
where σ, ρ, L, and U represent the droplet’s surface tension, density, characteristic length, and velocity, respectively. In this study, L and U are related to the LM volume and moving frequency, respectively.
If the effect of the inertia force versus the surface tension becomes large (We ↑), then the LM droplet cannot maintain its shape and hence exhibits various unexpected motions (e.g., oscillation–deformation, break up, etc.) [18]. These phenomena complicate the design of the accelerometer (Fig. 4d).
The volume and velocity of the LM affected the stability of our device. A large LM volume can increase sensitivity in low speed ranges but increases instability in high speed ranges. Therefore, an optimized volume that matches with the moving frequency (velocity) of the application is crucial.
Conclusion
In this paper, the spring constant in an accelerometer using a liquidtype proof mass was analyzed. The Laplace pressure induced a restoring force from a guiding channel and a LM droplet. In other words, the Laplace pressure can be considered as a spring in the device. In addition, sandblasting treatment was considered to reduce the damping effect. The outputs of the device were estimated using the base excitation mathematical model. In a dynamic situation, through an interesting phenomenon where the LM droplet vibrated at a natural vibration in a sandblastingtreated device, the characteristic area was determined. By matching the estimated outputs from the base excitation model and the oscillation experimental results, the spring and damping constants of the device were determined in a dynamic situation (for oscillations below 5 Hz). The change in the spring constant of the device in a dynamic situation and the effects of high surface roughness were analyzed. Finally, the effect of the LM droplet volume on the spring constant and device sensitivity was analyzed.
Availability of data and materials
The datasets and a movie supporting the conclusions of this article are included within the article.
Abbreviations
 LM:

liquid metal
 MEMS:

microelectromechanical system
 PCB:

printed circuit board
 PSA:

pressure sensitive adhesive
References
 1.
Shaeffer DK (2013) MEMS inertia sensors: a tutorial overview. IEEE Commun Mag 51(4):100–109. https://doi.org/10.1109/MCOM.2013.6495768
 2.
Li Q (2015) Fabrication of a high sensitivity MEMS accelerometer with symmetrical doublesided serpentine beammass structure. In: 2015 IEEE SENSORS, Busan, South Korea, 1–4 November 2015. https://doi.org/10.1109/icsens.2015.7370430
 3.
Zhou X (2015) Design and fabrication of a MEMS capacitive accelerometer with fully symmetrical doublesided Hshaped beam structure. Microelectron Eng 131:51–57. https://doi.org/10.1016/j.mee.2014.10.005
 4.
Sun CM (2008) On the sensitivity improvement of CMOS capacitive accelerometer. Sens Actuator APhys 141(2):347–352. https://doi.org/10.1016/j.sna.2007.10.026
 5.
Gomathi T (2016) Capacitive accelerometers for microelectromechanical applications: a review. In: 2016 ICCICCT, Kumaracoil, India, 16–17 December 2016. https://doi.org/10.1109/iccicct.2016.7987999
 6.
Mohammed Z (2018) Monolithic multi degree of freedom (MDoF) capacitive MEMS accelerometers. Micromachines 9(11):602. https://doi.org/10.3390/mi9110602
 7.
Partridge A (2000) A highperformance planar piezoresistive accelerometer. J Microelectromech Syst 9(1):58–66. https://doi.org/10.1109/84.825778
 8.
Hsieh HS (2011) A novel stress isolation guardring design for the improvement of a threeaxis piezoresistive accelerometer. In: 2011 16th International SolidState Sensors, Actuators and Microsystems Conference, Beijing, China, 5–9 June 2011. https://doi.org/10.1109/transducers.2011.5969582
 9.
Park U (2014) A micromachined differential resonant accelerometer based on robust structural design. Microelectron Eng 129:5–11. https://doi.org/10.1016/j.mee.2014.06.008
 10.
Zhang J (2015) Microelectromechanical resonant accelerometer designed with a high sensitivity. Sensors 15(12):30293–30310. https://doi.org/10.3390/s151229803
 11.
Krause AG (2012) A highresolution microchip optomechanical accelerometer. Nat Photonics 6:768–772. https://doi.org/10.1038/nphoton.2012.245
 12.
Cervantes FG (2014) High sensitivity optomechanical reference accelerometer over 10 kHz. Appl Phys Lett 104(22):221111. https://doi.org/10.1063/1.4881936
 13.
Langfelder G (2012) The dependence of fatigue in microelectromechanical systems on the environment and the industrial packaging. IEEE Trans Ind Electron 59(12):4938–4948. https://doi.org/10.1109/TIE.2011.2151824
 14.
Mariani S (2008) A threescale FE approach to reliability analysis of MEMS sensors subject to impacts. Meccanica 43(5):469–483. https://doi.org/10.1007/s1101200891110
 15.
Park U (2010) Development of a MEMS digital accelerometer (MDA) using a microscale liquid metal droplet in a microstructured photosensitive glass channel. Sens Actuator APhys 159(1):51–57. https://doi.org/10.1016/j.sna.2010.02.011
 16.
Huh M (2017) Simple and robust resistive dualaxis accelerometer using a liquid metal droplet. Micro Nano Syst Lett 5:5. https://doi.org/10.1186/s4048601600382
 17.
Won D (2020) Capacitivetype twoaxis accelerometer with liquidtype proof mass. Adv Electron Mater. https://doi.org/10.1002/aelm.201901265
 18.
Chen Y (2017) Aerodynamic breakup and secondary drop formation for a liquid metal column in a shockinduced crossflow. In: 55th AIAA Aerospace Sciences Meeting, AIAA, Grapevine, United States, 9–13 January 2017. https://doi.org/10.2514/6.20171892
Acknowledgements
Not applicable.
Funding
This work was supported by the National Research Foundation of Korea (NRF) Grant funded by the Korea government (MSIT) (No. 2017R1C1B5076710).
Author information
Affiliations
Contributions
DJW carried out the experiments and drafted the manuscript. SL analyzed the experimental result. JK drafted the manuscript. All authors read and approved the final manuscript.
Corresponding authors
Ethics declarations
Competing interests
The authors declare that they have no competing interests.
Additional information
Publisher's Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Supplementary information
Additional file 1: Movie S1.
The behavior of liquid metal droplets in the devices with (left) and without sandblasting treatments (right).
Rights and permissions
Open Access This article is licensed under a Creative Commons Attribution 4.0 International License, which permits use, sharing, adaptation, distribution and reproduction in any medium or format, as long as you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons licence, and indicate if changes were made. The images or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.
About this article
Cite this article
Won, D., Lee, S. & Kim, J. Analysis of liquidtype proof mass under oscillating conditions. Micro and Nano Syst Lett 8, 12 (2020). https://doi.org/10.1186/s4048602000115y
Received:
Accepted:
Published:
Keywords
 Liquid metal droplet
 Guiding channel
 Dynamic situation
 Surface modification
 Spring constant