- Open Access
Nonenzymatic coated screen-printed electrode for electrochemical determination of acetylcholine
© The Author(s) 2018
- Received: 18 June 2018
- Accepted: 9 November 2018
- Published: 26 November 2018
In the present study, a screen-printed electrode based-sensor with electrochemical detection was developed for rapid and sensitive determination of acetylcholine. At first, the screen-printed carbon electrode was modified by using a magnetic core shell and then was used for voltammetric determination of acetylcholine in ampoule, serum and urine samples. The electrochemical behaviour of acetylcholine at the modified electrode was investigated by cyclic voltammetry. The modified electrode displayed a decrease in the overpotential (ca. 130 mV) and an obvious increase in the peak current compared to the non-modified screen-printed electrode. The results indicated that modified screen-printed electrode enhanced electrocatalytic activity towards the oxidation of acetylcholine. Under optimized conditions, the limit of detection from the experiment of acetylcholine determination was 0.02 µM with acetylcholine concentration in range of 0.1–500.0 µM. The reproducibility of the measurements was tested by recording the responses for 50.0 µM acetylcholine with four different developed sensors prepared on the same manner and the same day. A relative standard deviation value of 4.3% was obtained. Finally, a recovery test is done as a part of accuracy evaluation of the system.
- Acetylcholine determination
- Magnetic core shell nanoparticles
- Screen-printed carbon electrode
The nervous system plays an important role in the human body and control body functions by regulating and coordinating their various activities, which is mainly carried out in the form of electrochemical signaling by neurotransmitters .
Acetylcholine (ACh) is an important neurotransmitter in both peripheral and central nervous systems. ACh have vital roles in the brain. ACh is implicated in many human behaviors, such as arousal and attention, and plays a key role in memory formation and learning. Decreases in the level of ACh cause various neurological disorders, including Parkinson’s disease, Alzheimer’s and dementia, and schizophrenia. On the other hand, increases in the level of ACh result in a decreased heart rate and increased production of saliva [2–6]. Hence, the quantitative determination of ACh is very important in biological sciences and clinical analysis.
Various methods had been reported for ACh detection which includes matrix-assisted laser desorption ionization time-of-flight mass spectrometry , high performance liquid chromatography coupled to post-column chemiluminescence detection , gas chromatography mass spectrometry , capillary zone electrophoresis , potentiometry , colorimetric detection , optical detection , chemiluminescence [14, 15], photoelectrochemistry  and enzymatic electrochemical biosensors . These methods are complicated, time-consuming and require expensive equipment that is not practical for widespread application.
Over the past decade, development of electrochemical non-enzymatic ACh biosensors has risen at considerable rate for ACh detection. The advantages include simple design, cost effectiveness, good stability and effective enzyme-like catalysis against temperature and pH .
Nanomaterials, because of their unique properties, have been extensively developed. Nanoparticles can act as conduction centers facilitating the transfer of electrons and provide great catalytic surface areas [19–23]. Among them, nanosized metal particle modified electrodes have emerged as a promising alternative for the electroanalysis of organic and inorganic compounds [24–28]. Metal nanoparticles have some distinct advantages such as higher mass transport, lower influence of the solution resistance, low detection limit, and better signal-to noise ratio over the conventional macroelectrodes [29–31].
The development of screen-printed electrodes (SPEs) has become a major revolution in the construction of electrochemical sensors/biosensors . The SPEs have been designed especially for miniaturization of electrochemical analytical systems . SPEs are highly-versatile, easy to use, cost-effective analytical tools, also suitable to miniaturization . Furthermore, a SPE avoids the cleaning process, unlike conventional electrodes such as a GCE .
In the present work, we synthesized magnetic core–shell manganese ferrite nanoparticles (MCSNP)  and screen printed carbon electrodes were modified with MCSNP. To the best of our knowledge, no study has been reported so far on the determination of acetylcholine by using MCSNP/SPE.
Apparatus and chemicals
The electrochemical measurements were performed with an Autolab potentiostat/galvanostat (PGSTAT 302 N, Eco Chemie, the Netherlands). The experimental conditions were controlled with General Purpose Electrochemical System software. Screen printed carbon electrodes were purchased from Drop Sens Co. A Metrohm 710 pH meter was used for pH measurements.
Acetylcholine chloride and all the other reagents were of analytical grade and were obtained from Merck (Darmstadt, Germany). The buffer solutions were prepared from orthophosphoric acid and its salts in the pH range of 2.0‒9.0. Magnetic core–shell manganese ferrite nanoparticles were synthesized in our laboratory as reported previously .
Preparation of the electrode
The bare screen-printed electrode was coated with MCSNP as follows. A stock solution of MCSNP in 1 mL aqueous solution was prepared by dispersing 1 mg MCSNP with ultrasonication for 1 h, and a 4 µL aliquot of the MCSNP/H2O suspension solution was casted on the carbon working electrodes, waiting until the solvent was evaporated in room temperature.
Preparation of serum, ampule and urine samples
The sample of the ACh ampule was prepared by the appropriate dilution with 0.1 M PBS solution (pH 7.0) and directly used for determination of ACh. Finally, a suitable volume of the resultant solutions were transfer to electrochemical cell and the resulting solution was used for the analysis of ACh.
Human serum samples were obtained from a Hospital in Kerman. The only pretreatment was ten-fold dilution of serum sample with buffer solution.
Urine samples were stored in a refrigerator immediately after collection. Twenty milliliters of the sample was centrifuged for 10 min at 3000 rpm. The supernatant was filtered out using a 0.45 µm filter. Then, different volume of the solution was transferred into a 50 mL volumetric flask and diluted to the mark with PBS (pH 7.0). The diluted urine sample was spiked with different amounts of ACh. The ACh contents were analyzed by the proposed method using the standard addition method in order to prevent any matrix effect.
The samples were spiked with different amounts of ACh and contents were analyzed by using the standard addition method in order to prevent any matrix effect. The amount of unknown ACh in the ACh ampule can be detected by extrapolating the plot.
Electrochemical behavior of acetylcholine at the surface MCSNP/SPE
The electrochemical behavior of ACh is dependent on the pH value of the aqueous solution. Therefore, pH optimization of the solution seems to be necessary in order to obtain the best results for electrooxidation of ACh. Thus the electrochemical behaviors of ACh were studied in 0.1 M PBS in different pH values (2.0–9.0) at the surface of MCSNP/SPE by voltammetry. It was found that the electro-oxidation of ACh at the surface of MCSNP/SPE was more favored under neutral conditions than in acidic or basic medium. Thus, the pH 7.0 was chosen as the optimum pH for electro-oxidation of ACh at the surface of MCSNP/SPE.
Also, MCSNP/SPE shows much higher anodic peak current for the oxidation of ACh compared to unmodified SPE, indicating that the modification of unmodified SPE with MCSNP has significantly improved the performance of the electrode toward ACh oxidation.
Effect of scan rate
Calibration plots and limit of detection
Effect of interference in the determination of 50.0 μM ACh (equal molar)
Current in the absence of interference (μA)
Current in the presence of interference (μA)
Glucose, l-asparagine, saccarose, isoproterenol
l-Proline, benzoic acid
Fructose, ascorbic acid
Methanol, serotonin, Fe+3
The repeatability and stability of the MCSNP/SPE
The long-term stability of the MCSNP/SPE was tested over a 3 week period that stored in atmosphere at room temperature. Then, CVs were recorded. The results were showed that the peak potential for ACh oxidation was unchanged and the current signals showed less than 2.6% decrease relative to the initial response. The antifouling properties of the MCSNP/SPE toward ACh oxidation and its oxidation product was investigated by recording the CVs of the MCSNP/SPE before and after use in the presence of ACh. CVs were recorded in the presence of ACh after having cycled the potential 15 times at a scan rate of 50 mV/s. The results were showed that the peak potentials were unchanged and the currents decreased by less than 2.3%. Therefore, on the surface of the MCSNP/SPE, not only does the sensitivity increase, but the fouling effect of the analyte and its oxidation product also decreases.
ACh ampule, urine and serum analysis
The application of MCSNP/SPE for determination of ACh in ampoule, serum and urine samples (n = 5)
The amounts of ACh in ampule was found to be 50.07 mg/mL. It was found that there is no significant difference between the results obtained by the MCSNP/SPE and the nominal value on the ampoule label (50.00 mg/mL). The t-test was applied to both sets of results and showed that there was no significant difference at the 95% confidence level.
In this work, employing magnetic core shell nanoparticles as modifier in modification of SPEs, a novel sensor has been developed that provides a sensitive method for the determination of ACh. The proposed protocol demonstrated herein a novel, simple, portable, inexpensive and easy-to-use fabrication method for the measurement of ACh concentration in ampoule, serum and urine samples with good analytical performance. Due to the unique properties of magnetic core shell nanoparticles, the sensor exhibited remarkable electrochemical activity toward the oxidation of ACh. Under optimized conditions, DPV exhibited linear dynamic ranges from 0.1 to 500 µM with detection limit of 20.0 nM.
SZM and HB conceived experiments and discussed results. ST and HB designed assays and performed experiments. SZM and HB analysed data. SZM wrote the manuscript with input from ST and HB. All authors read and approved the final manuscript.
The authors wish to thank Payame Noor University for support of this work.
The authors declare that they have no competing interests.
Availability of data and materials
All data generated or analysed during this study are included in this published article.
This work was supported by Payame Noor University (Grant number: 968532).
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Chauhan N, Pundir CS (2014) Amperometric determination of acetylcholine—a neurotransmitter, by chitosan/gold-coated ferric oxide nanoparticles modified gold electrode. Biosens Bioelectron 61:1–8View ArticleGoogle Scholar
- Bolat EO, Tıg GA, Pekyardımcı S (2017) Fabrication of an amperometric acetylcholine esterase-choline oxidase biosensor based on MWCNTs-Fe3O4NPs-CS nanocomposite for determination of acetylcholine. J Electroanal Chem 785:241–248View ArticleGoogle Scholar
- Yang M, Yang Y, Yang Y, Shen G, Yu R (2005) Microbiosensor for acetylcholine and choline based on electropolymerization/sol–gel derived composite membrane. Anal Chim Acta 530:205–211View ArticleGoogle Scholar
- Xue W, Cui T (2008) A thin-film transistor based acetylcholine sensor using self-assembled carbon nanotubes and SiO2 nanoparticles. Sens Actuators B 134:981–987View ArticleGoogle Scholar
- Çevik S, Timur S, Anik U (2012) Biocentri-voltammetric biosensor for acetylcholine and choline. Microchim Acta 179:299–305View ArticleGoogle Scholar
- Shimomura T, Itoh T, Sumiya T, Mizukami F, Ono M (2009) Amperometric biosensor based on enzymes immobilized in hybrid mesoporous membranes for the determination of acetylcholine. Enzyme Microb Technol 45:443–448View ArticleGoogle Scholar
- Persike M, Zimmermann M, Klein J, Karas M (2010) Quantitative determination of acetylcholine and choline in microdialysis samples by MALDI-TOF MS. Anal Chem 82:922–929View ArticleGoogle Scholar
- Yoshida H, Yamada A, Todoroki K, Imakyure O, Nohta H, Yamaguchi M (2009) Liquid chromatographic determination of acetylcholine based on pre-column alkaline cleavage reaction and post-column tris (2,2′-bipyridyl) ruthenium (III) chemiluminescence detection. Luminescence 24:306–310View ArticleGoogle Scholar
- Patterson TA, Kosh JW (1992) Simultaneous quantitation of arecoline, acetylcholine, and choline in tissue using gas chromatography/electron impact mass spectrometry. Biol Mass Spectrom 21:299–304View ArticleGoogle Scholar
- Carter N, Trenerry VC (1996) The determination of choline in vitamin preparations, infant formula and selected foods by capillary zone electrophoresis with indirect ultraviolet detection. Electrophoresis 17:1622–1626View ArticleGoogle Scholar
- Barsoum BN, Watson WM, Mahdi IM, Khalid E (2004) Electrometric assay for the determination of acetylcholine using a sensitive sensor based on carbon paste. J Electroanal Chem 567:277–281View ArticleGoogle Scholar
- Qian J, Yang X, Jiang L, Zhu C, Mao H, Wang K (2014) Facile preparation of Fe3O4 nanospheres/reduced graphene oxide nanocomposites with high peroxidase-like activity for sensitive and selective colorimetric detection of acetylcholine. Sens Actuators B Chem 201:160–166View ArticleGoogle Scholar
- Wang CI, Periasamy AP, Chang HT (2013) Photoluminescentc-dots@RGO probe for sensitive and selective detection of acetylcholine. Anal Chem 85:3263–3270View ArticleGoogle Scholar
- Buiculescu R, Hatzimarinaki M, Chaniotakis N (2010) Biosilicated CdSe/ZnS quantum dots as photoluminescent transducers for acetylcholinesterase-based biosensors. Anal Bioanal Chem 398:3015–3021View ArticleGoogle Scholar
- Burmeister JJ, Pomerleau F, Huettl P, Gash CR, Werner CE, Bruno JP, Gerhardt GA (2008) Ceramic-based multisite microelectrode arrays for simultaneous measures of choline and acetylcholine in CNS. Biosens Bioelectron 23:1382–1389View ArticleGoogle Scholar
- Zayats M, Kharitonov AB, Pogorelova SP, Lioubashevski O, Katz E, Willner I (2003) Probing photoelectrochemical processes in Au-CdS nanoparticle arrays by surface plasmon resonance: application for the detection of acetylcholine esterase inhibitors. J Am Chem Soc 125:16006–16014View ArticleGoogle Scholar
- Zhu W, An YR, Zheng JH, Tang LL, Zhang W, Jin LT, Jiang L (2009) A new microdialysis electrochemical device for in vivo simultaneous determination of acetyl-choline and choline in rat brain treated with N-methyl-(R)-salsolinol. Biosens Bioelectron 24:3594–3599View ArticleGoogle Scholar
- Anithaa AC, Asokan K, Sekar C (2018) Low energy nitrogen ion beam implanted tungsten trioxide thin films modified indium tin oxide electrode based acetylcholine sensor. J Taiwan Inst Chem Eng 84:11–18View ArticleGoogle Scholar
- Mohammadi SZ, Beitollahi H, Jasemi M, Akbari A (2015) Nanomolar determination of methyldopa in the presence of large amounts of hydrochlorothiazide using a carbon paste electrode modified with graphene oxide nanosheets and 3-(4′-amino-3′-hydroxy-biphenyl-4-yl)acrylic acid. Electroanalysis 27:2421–2430View ArticleGoogle Scholar
- Beitollahi H, Ghofrani Ivari S, Torkzadeh Mahani M (2016) Voltammetric determination of 6-thioguanine and folic acid using a carbon paste electrode modified with ZnO-CuO nanoplates and modifier. Mater Sci Eng C 69:128–133View ArticleGoogle Scholar
- Mohammadi SZ, Beitollahi H, Hassanzadeh M (2018) Voltammetric determination of tryptophan using a carbon paste electrode modified with magnesium core shell nanocomposite and ionic liquids. Anal Bioanal Chem Res 5:55–65Google Scholar
- Jahani Sh, Beitollahi H (2016) Selective detection of dopamine in the presence of uric acid using NiO nanoparticles decorated on graphene nanosheets modified screen-printed electrodes. Electroanalysis 28:2022–2028View ArticleGoogle Scholar
- Mohammadi SZ, Beitollahi H, Bani Asadi E (2015) Electrochemical determination of hydrazine using a ZrO2 nanoparticles-modified carbon paste electrode. Environ Monit Assess 187:122–132View ArticleGoogle Scholar
- Mohammadi SZ, Beitollahi H, Nikpour N, Hosseinzadeh R (2016) Electrochemical sensor for determination of ascorbic acid using a 2-chlorobenzoyl ferrocene/carbon nanotube paste electrode. Anal Bioanal Chem Res 3:187–194Google Scholar
- Pingarrón JM, Yanez-Sedeno P, lez-Cortes AG (2008) Gold nanoparticle-based electrochemical biosensors. Electrochim Acta 53:5848–5866View ArticleGoogle Scholar
- Afkhami A, Gomar F, Madrakian T (2016) CoFe2O4 nanoparticles modified carbon paste electrode for simultaneous detection of oxycodone and codeine in human plasma and urine. Sens Actuators B 233:263–271View ArticleGoogle Scholar
- Zhang Y, Suryanarayanan V, Nakazawa I, Yoshihara S, Shirakashi T (2004) Electrochemical behavior of Au nanoparticle deposited on as-grown and O-terminated diamond electrodes for oxygen reduction in alkaline solution. Electrochim Acta 49:5235–5240View ArticleGoogle Scholar
- Beitollahi H, Mohammadi S (2013) Selective voltammetric determination of norepinephrine in the presence of acetaminophen and tryptophan on the surface of a modified carbon nanotube paste electrode. Mater Sci Eng C 33:3214–3219View ArticleGoogle Scholar
- Penner RM, Martin CR (1987) Preparation and electrochemical characterization of ultramicroelectrode ensembles. Anal Chem 59:2625–2630View ArticleGoogle Scholar
- Reller H, Kirowa-Eisner E, Gileadi E (1984) Ensembles of microelectrodes: digital simulation by the two-dimensional expanding grid method: cyclic voltammetry, IR effects and applications. J Electroanal Chem 161:247–268View ArticleGoogle Scholar
- Cassidy J, Ghoroghchian J, Sarfarazi F, Smith JJ, Pons S (1986) Simulation of edge effects in electroanalytical experiments by orthogonal collocation—VI. Cyclic voltammetry at ultramicroelectrode ensembles. Electrochim Acta 31:629–636View ArticleGoogle Scholar
- Cumba LR, Smith JP, Zuway KY, Sutcliffe OB, do Carmo DR, Banks CE (2016) Forensic electrochemistry: simultaneous voltammetric detection of MDMA and its fatal counterpart “Dr Death” (PMA). Anal Methods 8:142–152View ArticleGoogle Scholar
- Chan KF, Lim HN, Shams N, Jayabal S, Pandikumar A, Huang NM (2016) Fabrication of graphene/gold-modified screen-printed electrode for detection of carcinoembryonic antigen. Mater Sci Eng C 58:666–674View ArticleGoogle Scholar
- Chatzipetrou M, Milano F, Giotta L, Chirizzi D, Trotta M, Massaouti M, Guascito MR, Zergioti I (2016) Functionalization of gold screen printed electrodes with bacterial photosynthetic reaction centers by laser printing technology for mediatorless herbicide biosensing. Electrochem Commun 64:46–50View ArticleGoogle Scholar
- Lezi N, Economou A, Barek J, Prodromidis M (2014) Screen-printed disposable sensors modified with bismuth precursors for rapid voltammetric determination of 3 ecotoxic nitrophenols. Electroanalysis 26:766–775View ArticleGoogle Scholar
- Mohammadi SZ, Seyedi A (2016) Preconcentration of cadmium and copper ions on magnetic core–shell nanoparticles for determination by flame atomic absorption. Toxicol Environ Chem 98:705–713Google Scholar
- Bard AJ, Faulkner LR (2001) Electrochemical methods: fundamentals and applications, 2nd edn. Wiley, New YorkGoogle Scholar