- Open Access
Feasibility study of a biocompatible pneumatic dispensing system using mouse 3T3-J2 fibroblasts
© The Author(s) 2017
- Received: 25 September 2017
- Accepted: 20 November 2017
- Published: 24 November 2017
This paper presents results for dispensing living cells using a pneumatic dispensing system to verify the feasibility of using this system to fabricate biomaterials. Living cells (i.e., mouse 3T3-J2 fibroblast) were dispensed with different dispensing pressures in order to evaluate the effect of dispensing process on cell viability and proliferation. Based on the results of a live-dead assay, more than 80% of cell viability has been confirmed which was reasonably similar to that in the control group. Furthermore, measurement of cell metabolic activity after dispensing confirmed that the dispensed cell proliferated at a rate comparable to that of the control group. These results demonstrate that the pneumatic dispensing system is a promising tool for fabrication of biomaterials.
- Pneumatic dispensing
- Living cell
Technologies to manipulate living cells are widely employed in biological and tissue engineering applications. For example, depositing living cells at a specific location is useful to help understand how cell behavior such as proliferation, differentiation, and migration is related to local environment (e.g., extracellular matrix and neighboring cells) [1–4]. Also, in tissue engineering, which seeks to repair injured or ill organs, it is necessary to seed living cells at a precise position in three-dimensional (3D) scaffolds to achieve successful production of artificial tissues or organs [5–7].
To construct 3D cellular structures, many cell patterning technologies have been developed, including micro-contact printing using soft-lithography , and microfluidic technology . These technologies can produce a high resolution pattern, but they are inconvenient to form 3D structures and expensive to change patterns .
Recently, dispensing (e.g., inkjet printing) technology has attracted much attention for use as a tool to fabricate structures consisting of various cell types [11–15]. Because the dispensing technology can make a precise cell pattern directly by dispensing small droplets of biological materials containing living cells, it has advantages of simplicity and flexibility for fabrication of complex cellular structures. The dispensing technology is flexible to pattern various designs by simply controlling the position to dispense living cells in droplets from a nozzle. Furthermore, because the dispensing technology can be easily integrated with computer-assisted manufacturing systems, it can use solid free-form fabrication to precisely form complex 3D cellular structures [16, 17]. Also, it can form cellular structures that consist of different cell types by using and controlling multiple nozzles, each with a corresponding reservoir .
For tissue engineering applications, considerable research has been devoted to extending the capability of dispensing technology to create artificial tissues or organs. Because living cells are easily damaged by heat and mechanical stress , the biocompatibility of the dispensing system must be assessed by evaluating the effect of the dispensing process on cell viability and proliferation. Living cells have been dispensed using various printing systems such as thermal inkjets , piezoelectric inkjets [12–15], electrostatics , lasers , and electrohydrodynamic jets .
This paper presents results for dispensing living cells using a pneumatic dispensing system to verify the feasibility of using it as a biocompatible fabrication tool. The pneumatic dispensing system uses a simple yet effective mechanism with a backflow stopper and a flexible membrane, so it is easy to control droplet volumes and to eject highly viscous liquid [20, 21]. To assess the biocompatibility of the dispensing system, experiments were conducted using mouse 3T3-J2 fibroblast cells at various operation conditions, then cell viability and proliferation were evaluated using a live-dead assay and a cell counting kit assay, respectively.
Pneumatic cell dispensing system
Summary of design parameters
The flexible membrane is deflected by the applied pressure, and thus draws in or dispenses liquid. The membrane is either pulled (negative pressure) or pushed (positive pressure) depending on the programmed electric signal. The applied pressure is normally negative; a pulsed signal switches a solenoid valve to provide positive pressure (during duration time) to dispense the liquid. The liquid is drawn into the chamber when the membrane is pulled (during delay time) and dispensed when it is pushed.
Preparation of cell suspension
A cell suspension was prepared using mouse fibroblasts (3T3-J2 cells) obtained from the American Type Culture Collection (Manassas, VA). Cells were cultured in Dulbecco’s modified Eagle’s medium (DMEM, Gibco BRL, Gaithersburg, MD) supplemented with 10% bovine calf serum (Gibco), 1% mixture of penicillin and streptomycin (Gibco). Cells were cultured at 37 °C in a humidified incubator in an atmosphere containing 10% CO2 . To apply living cells to the dispensing system, cells in culture flasks were trypsinized, then cell pellets were collected, and resuspended in a phosphate-buffered saline (PBS, Gibco) solution. The suspension contained 500,000 cells mL−1, as quantified using a hemocytometer.
Cell viability assay
Cell proliferation assay
To test the biocompatibility of our dispensing system, cell viability and proliferation were analyzed qualitatively using live-dead and CCK-8 assays. A statistical significance was determined using analysis of variance (ANOVA) on MINITAB version 14.2 (Minitab Inc., State College, PA, USA). A P value less than 0.05 was considered statistically significant.
The effect of our dispensing system on cell viability and proliferation was assessed at various applied pressures. Based on the results of a live-dead assay, more than 80% of cell viability has been confirmed which is reasonably compatible to the control group. Cell metabolic activity measurements confirmed that the dispensed cells were proliferating at a rate comparable to that of the control. These results confirm the feasibility of using our pneumatic dispensing system to dispense living cells for fabrication of biomaterials.
LS and KH performed the experiments, analyzed the data and wrote the manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Availability of data and materials
The datasets supporting the conclusions of this article are included within the article and its additional files.
Ethics approval and consent to participate
This work was supported by Dong-eui University (Grant No. 201702730001) and the National Research Foundation of Korea (NRF) Grant funded by the Korea government (MSIT) (Nos. 2017R1C1B1008045 and 2017R1C1B5076710).
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Devillard R, Pagès E, Correa MM, Kériquel V, Rémy M, Kalisky J, Ali M, Guillotin B, Guillemot F (2014) Cell Patterning by laser-assisted bioprinting. Methods Cell Biol 119:159–174. https://doi.org/10.1016/B978-0-12-416742-1.00009-3 View ArticleGoogle Scholar
- Shi M, Ling K, Yong KW, Li Y, Feng S, Zhang X, Pingguan-Murphy B, Lu TJ, Xu F (2015) High-throughput non-contact vitrification of cell-laden droplets based on cell printing. Sci Rep 5:17928. https://doi.org/10.1038/srep17928 View ArticleGoogle Scholar
- Xu T, Gregory CA, Molnar P, Cui X, Jalota S, Bhaduri SB, Boland T (2006) Viability and electrophysiology of neural cell structures generated by the inkjet printing method. Biomaterials 27:3580–3588Google Scholar
- Miller ED, Fisher GW, Weiss LE, Walker LM, Campbell PG (2006) Dose-dependent cell growth in response to concentration modulated patterns of FGF-2 printed on fibrin. Biomaterials 27:2213–2221View ArticleGoogle Scholar
- Nakamura M, Kobayashi A, Takagi F, Watanabe A, Hiruma Y, Ohuchi K, Iwasaki Y, Horie M, Morita I, Takatani S (2005) Biocompatible inkjet printing technique for designed seeding of individual living cells. Tissue Eng 11:1658–1666View ArticleGoogle Scholar
- Nakamura M, Iwanaga S, Henmi C, Arai K, Nishiyama Y (2010) Biomatrices and biomaterials for future developments of bioprinting and biofabrication. Biofabrication 2:014110. https://doi.org/10.1088/1758-5082/2/1/014110 View ArticleGoogle Scholar
- Matsusaki M, Sakaue K, Kadowaki K, Akashi M (2013) Three-dimensional human tissue chips fabricated by rapid and automatic inkjet cell printing. Adv Healthc Mater 2:534–539View ArticleGoogle Scholar
- Tanaka N, Ota H, Fukumori K, Miyake J, Yamato M, Okano T (2014) Micro-patterned cell-sheets fabricated with stamping-force-controlled micro-contact printing. Biomaterials 35(37):9802–9810View ArticleGoogle Scholar
- Collins DJ, Morahan B, Garcia-Bustos J, Doerig C, Plebanski M, Neild A (2015) Two-dimensional single-cell patterning with one cell per well driven by surface acoustic waves. Nat Commun 6:8686. https://doi.org/10.1038/ncomms9686 View ArticleGoogle Scholar
- Goubko CA, Cao X (2009) Patterning multiple cell types in co-cultures: a review. Mater Sci Eng, C 29:1855–1868View ArticleGoogle Scholar
- Xu T, Jin J, Gregory C, Hickman JJ, Boland T (2005) Inkjet printing of viable mammalian cells. Biomaterials 26:93–99View ArticleGoogle Scholar
- Lorber B, Hsiao W-K, Hutchings IM, Martin KR (2013) Adult rat retinal ganglion cells and glia can be printed by piezoelectric inkjet printing. Biofabrication 6(1):015001. https://doi.org/10.1088/1758-5082/6/1/015001 View ArticleGoogle Scholar
- Zhang J, Chen F, He Z, Ma Y, Uchiyama K, Lin J-M (2016) A novel approach for precisely controlled multiple cell patterning in microfluidic chips by inkjet printing and the detection of drug metabolism and diffusion. Analyst 141:2940. https://doi.org/10.1039/c6an00395h View ArticleGoogle Scholar
- Xu T, Zhao W, Zhu J-M, Albanna MZ, Yoo JJ, Atala A (2013) Complex heterogeneous tissue constructs containing multiple cell types prepared by inkjet printing technology. Biomaterials 34:130–139View ArticleGoogle Scholar
- Kim YK, Park JA, Yoon WH, Kim J, Jung S (2016) Drop-on-demand inkjet-based cell printing with 30-μm nozzle diameter for cell-level accuracy. Biomicrofluidics 10:064110View ArticleGoogle Scholar
- Kim JY, Yoon JJ, Park EK, Kim DS, Kim S-Y, Cho D-W (2009) Cell adhesion and proliferation evaluation of SFF-based biodegradable scaffolds fabricated using a multi-head deposition system. Biofabrication 1(1):015002View ArticleGoogle Scholar
- Mironov V, Boland T, Trusk T, Forgacs G, Markwald RR (2003) Organ printing: computer-aided jet-based 3D tissue engineering. Trends Biotechnol 21(4):157–161View ArticleGoogle Scholar
- Xiong R, Christensen K, Fu J, Markwald RR, Huang Y (2016) Freeform laser and inkjet printing of biological constructs. 2016 International symposium on flexible automation, Cleveland, Ohio, USA, 1–3 August 2016Google Scholar
- Liaudanskaya V, Gasperini L, Maniglio D, Motta A, Migliaresi C (2015) Assessing the impact of electrohydrodynamic jetting on encapsulated cell viability, proliferation, and ability to self-assemble in three-dimensional structures. Tissue Eng Part C 21(6):631–638. https://doi.org/10.1089/ten.tec.2014.0228 View ArticleGoogle Scholar
- Lee S, Kim J (2010) Development and characterization of a cartridge-type pneumatic dispenser with an integrated backflow stopper. J Micromech Microeng 20(1):015011. https://doi.org/10.1088/0960-1317/20/1/015011 View ArticleGoogle Scholar
- In Ho Choi and Joonwon Kim (2016) A pneumatically driven inkjet printing system for highly viscous microdroplet formation. Micro Nano Syst Lett 4:4. https://doi.org/10.1186/s40486-016-0030-x View ArticleGoogle Scholar
- Park J, Berthiaume F, Toner M, Yarmush ML, Tilles AW (2005) Microfabricated grooved substructures as platforms for bioartificial liver reactors. Biotechnol Bioeng 90:632–644View ArticleGoogle Scholar
- Liberski AR, Zhang R, Bradley M (2009) In situ nanoliter-scale polymer fabrication for flexible cell patterning. JALA 14:285–293Google Scholar
- Lee S, Choi IH, Kim YK, Kim J (2014) Velocity control of nanoliter droplets using a pneumatic dispensing system. Micro Nano Syst Lett 2:5. https://doi.org/10.1186/s40486-014-0005-8 View ArticleGoogle Scholar
- Saunders RE, Gough JE, Derby B (2008) Delivery of human fibroblast cells by piezoelectric drop-on-demand inkjet printing. Biomaterials 29:193–203View ArticleGoogle Scholar