The potential of a dielectrophoresis activated cell sorter (DACS) as a next generation cell sorter
© The Author(s). 2016
Received: 7 March 2016
Accepted: 10 May 2016
Published: 20 May 2016
Originally introduced by H. A. Pohl in 1951, dielectrophoretic (DEP) force has been used as a striking tool for biological particle manipulation (or separation) for the last few decades. In particular, dielectrophoresis activated cell sorters (DACSes) have been developed for applications in various biomedical fields. These applications include cell replacement therapy, drug screening and medical diagnostics. Since a DACS does not require a specific bio-marker, it is able to function as a biological particle sorting tool with numerous configurations for various cells [e.g. red blood cells (RBCs), white blood cells (WBCs), circulating tumor cells, leukemia cells, breast cancer cells, bacterial cells, yeast cells and sperm cells]. This article explores current DACS capabilities worldwide, and it also looks at recent developments intended to overcome particular limitations. First, the basic theories are reviewed. Then, representative DACSes based on DEP trapping, traveling wave DEP systems, DEP field-flow fractionation and DEP barriers are introduced, and the strong and weak points of each DACS are discussed. Finally, for the purposes of commercialization, prerequisites regarding throughput, efficiency and recovery rates are discussed in detail through comparisons with commercial cell sorters (e.g. fluorescent activated and magnetic activated cell sorters).
KeywordsDielectrophoresis Dielectrophoretic force (DEP force) Dielectrophoresis activated cell sorter (DACS) Microfluidics Cell sorting
Since dielectrophoresis was first reported by H. A. Pohl in 1951, it has been employed as a biological particle manipulating tool in various fields (e.g. cell replacement therapy, drug screening, medical diagnostics, particle filtration and microfluidics) [1–10]. Dielectrophoretic (DEP) phenomena occur when micro/nano-particles in a medium are exposed to a non-uniform electric field, causing polarization for particular particles according to their dielectric property [11, 12]. DEP force is classified into two types according to correlations of the dielectric properties of the particles and medium. Positive DEP (p-DEP) force pulls particles toward a higher electric gradient, and negative DEP (n-DEP) force repels particles away from the higher electric gradient. Therefore, various target cells with different dielectric properties can be manipulated by controlling the medium properties or the input voltage condition. In order to manipulate biological particles, consequently, various dielectrophoresis-based techniques, including DEP trapping, DEP field-flow fractionation (DEP-FFF), traveling wave DEP (TwDEP) force and DEP barrier, are performed within a micro fluid channel [13–21]. DEP trapping techniques are mainly used to isolate particles within a still fluid utilizing p-DEP force [22–25]. The TwDEP force, DEP-FFF and DEP barrier techniques are realized via angled or vertical electrode pairs, and they are generally implemented with n-DEP force within the micro channel with fluidic flow [26–32]. The latter techniques have a striking advantage in terms of throughput since the continuous loading of target cells along the fluidic flow allows for continuous cell separation. Therefore, there have been many studies on the separation of various biological particles, including blood cells [red blood cells (RBCs) and white blood cells (WBCs)], cancer cells [circulating tumor cells (CTCs), leukemia cells and breast cancer cells], submicron particles (bacteria, yeast cells), spermatozoa, stem cells, protein and DNA. Dielectrophoresis activated cell sorters (DACSes) differ from conventional fluorescent activated cell sorters (FACSes) and magnetic activated cell sorters (MACSes) in that they do not require the additional financial and time expenditure necessary for immune-labeling [33–38]. In addition, DACSes achieve high separation efficiency owing to the continuous separation in the micro fluid channel. Nevertheless, there are still big hurdles to be overcome with respect to their low throughputs and recovery rates.
In this article, therefore, we explore current DACS capabilities worldwide and look at recent developments intended to overcome particular limitations. First, the basic theories are reviewed. Then, the configurations and characteristics of four representative DACSes are compared, and the strong and weak points of each DACS are discussed. Finally, a commercialization strategy is suggested.
Traveling wave dielectrophoresis
DEP field flow fractionation
Comparative analysis with a commercial cell sorter
FACSes and MACSes are representative commercial cell sorters. FACSes show high separation efficiencies of over 97 %, throughputs of 10,000 cells/s and recovery rates of over 55 % [55–57]. MACSes show similar performance characteristics—separation efficiencies of over 90 %, throughputs of 1010 cells/h and recovery rates near 50 % [58–60]. The results do not include the time-consuming labeling process that takes over an hour. The aforementioned DACS employing DEP-FFF and DEP barrier immediately achieved competitive results when compared to FACSes and MACSes in terms of separation efficiency. However, the throughput and recovery rate are still not at commercial levels. Even though Hu et al.  reported a striking throughput employing dielectrophoresis at over 10,000 cells/s, similar to that of a FACS, their sorter required a specific marker and a labeling process as prerequisites, which are also required for FACSes and MACSes. Consequently, it can be said that a DACS without immune-labelling cannot produce the levels of performance in terms of efficiency, throughput and recovery rate when compared to FACSes and MACSes. Nevertheless, a DACS employing the intrinsic dielectric material properties of a particle obtained a high separation efficiency and high recovery rate without the labeling process. Once a high throughput can be achieved through an optimization process (e.g. employing a modular system consisting of a few of the same features as those used by MACSes, applying a higher flow rate and so on), DACSes will be sufficiently competitive with FACSes and MACSes.
Conclusion and prospects
Tissue engineering to rebuild damaged organs.
Stem cell separation for cell replacement treatment.
In vitro fertilization or intracytoplasmic sperm injection; superior sperm selection.
DL carried out the research work and wrote the manuscript. BH carried out a survey of dielectrohporesis-based particle sorting systems. BK supervised all research work. All authors read and approved the final manuscript.
Dongkyu Lee received his B.S. and M.S. degrees from Korea Aerospace University in 2012 and 2014, respectively, and is now a Ph.D. candidate at Korea Aerospace University. His current research interests include microelectromechanical systems (MEMS) for bio particle sorting systems.
Bohyun Hwang received his B.S. degree from Korea Aerospace University in 2015, and is now in a Master’s course at Korea Aerospace University. His current research interests include the development of cell sorting systems based on dielectrophoresis.
Byungkyu Kim received his Ph.D. in mechanical engineering from the University of Wisconsin, Madison, in 1997. From 1997 to 2000, he was a technical staff member at the Center for X-ray Lithography (CXrL) at the University of Wisconsin where he developed a computer code for the thermal modeling of a mask membrane and wafer during beam exposure. From 2000 to 2005, he worked for the Microsystem Center at KIST as a principal research scientist. He was in charge of developing the microrobot for a microcapsule-type endoscope. Currently, he is a professor in the School of Aerospace and Mechanical Engineering at Korea Aerospace University. His research interests include microelectromechanical systems (MEMS) for cell separation.
This research was supported by the Bio & Medical Technology Development Program (NRF-2005-2000206) and the Basic Science Research Program (NRF-2015R1D1A01057714) of the National Research Foundation of Korea (NRF) funded by the Ministry of Education, Science and Technology.
The authors declare that they have no competing interests.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Tsutsui Hideaki, Ho Chih-Ming (2009) Cell separation by non-inertial force fields in microfluidic systems. Mech Res Commun 36(1):92–103View ArticleMATHGoogle Scholar
- Franke Thomas A, Wixforth Achim (2008) Microfluidics for miniaturized laboratories on a chip. ChemPhysChem 9(15):2140–2156View ArticleGoogle Scholar
- Kersaudy-Kerhoas M, Dhariwal R, Desmulliez MPY (2008) Recent advances in microparticle continuous separation. IET Nanobiotechnol 2(1):1–13View ArticleGoogle Scholar
- Khoshmanesh Khashayar et al (2011) “Dielectrophoretic platforms for bio-microfluidic systems. Biosens Bioelectron 26(5):1800–1814View ArticleGoogle Scholar
- Zhang C et al (2010) Dielectrophoresis for manipulation of micro/nano particles in microfluidic systems. Anal Bioanal Chem 396(1):401–420View ArticleGoogle Scholar
- Qian Cheng et al (2014) “Dielectrophoresis for bioparticle manipulation. Int J Mol Sci 15(10):18281–18309View ArticleGoogle Scholar
- Lapizco-Encinas Blanca H, Rito-Palomares Marco (2007) Dielectrophoresis for the manipulation of nanobioparticles. Electrophoresis 28(24):4521–4538View ArticleGoogle Scholar
- Gascoyne Peter RC, Vykoukal Jody (2002) Particle separation by dielectrophoresis. Electrophoresis 23(13):1973View ArticleGoogle Scholar
- Kadaksham J, Singh P, Aubry N (2003) Particle separation using dielectrophoresis. ASME 2003 International Mechanical Engineering Congress and Exposition. American Society of Mechanical Engineers
- Hawkins BG et al (2007) Continuous-flow particle separation by 3D insulative dielectrophoresis using coherently shaped, dc-biased, ac electric fields. Anal Chem 79(19):7291–7300View ArticleGoogle Scholar
- Pohl Herbert A (1951) The motion and precipitation of suspensoids in divergent electric fields. J Appl Phys 22(7):869–871View ArticleGoogle Scholar
- Pohl HA, Pohl HA (1978) Dielectrophoresis: the behavior of neutral matter in nonuniform electric fields, vol 80. Cambridge University Press, CambridgeGoogle Scholar
- Suehiro Junya et al (2003) Selective detection of specific bacteria using dielectrophoretic impedance measurement method combined with an antigen–antibody reaction. J Electrost 58(3):229–246View ArticleGoogle Scholar
- Grilli Simonetta, Ferraro Pietro (2008) Dielectrophoretic trapping of suspended particles by selective pyroelectric effect in lithium niobate crystals. Appl Phys Lett 92(23):232902View ArticleGoogle Scholar
- Müller T et al (1996) High frequency electric fields for trapping of viruses. Biotechnol Tech 10(4):221–226View ArticleGoogle Scholar
- Morgan H et al (1997) Large-area travelling-wave dielectrophoresis particle separator. J Micromech Microeng 7(2):65View ArticleGoogle Scholar
- Choi Eunpyo, Kim Byungkyu, Park Jungyul (2009) High-throughput microparticle separation using gradient traveling wave dielectrophoresis. J Micromech Microeng 19(12):125014View ArticleGoogle Scholar
- Markx Gerard H, Rousselet Juliette, Pethig Ronald (1997) DEP-FFF: field-flow fractionation using non-uniform electric fields. J Liq Chromatogr Relat Technol 20(16-17):2857–2872View ArticleGoogle Scholar
- Dürr M et al (2003) Microdevices for manipulation and accumulation of micro-and nanoparticles by dielectrophoresis. Electrophoresis 24(4):722–731View ArticleGoogle Scholar
- Kentsch J et al (2003) Microdevices for separation, accumulation, and analysis of biological micro-and nanoparticles. IEE Proceedings-Nanobiotechnology. vol 150. No. 2. IET Digital Library
- Chen DF, Du H, Li WH (2006) A 3D paired microelectrode array for accumulation and separation of microparticles. J Micromech Microeng 16(7):1162View ArticleGoogle Scholar
- Müller T et al (1995) High-frequency electric-field trap for micron and submicron particles. Il Nuovo Cimento D 17(4):425–432View ArticleGoogle Scholar
- Thomas Rupert S, Morgan Hywel, Green Nicolas G (2009) Negative DEP traps for single cell immobilisation. Lab Chip 9(11):1534–1540View ArticleGoogle Scholar
- Zhao Y, Yi UC, Cho SK (2007) Microparticle concentration and separation by traveling-wave dielectrophoresis (twDEP) for digital microfluidics. Microelectromech Syst J 16(6):1472–1481View ArticleGoogle Scholar
- Zhao Y, Yi UC, Cho SK (2007) Highly efficient in-droplet particle concentration and separation by twDEP and EWOD for digital microfluidics. Micro Electro Mech Syst, MEMS. IEEE 20th International Conference on. IEEE
- Choi Sungyoung, Park Je-Kyun (2007) Continuous hydrophoretic separation and sizing of microparticles using slanted obstacles in a microchannel. Lab Chip 7(7):890–897View ArticleGoogle Scholar
- Huang Y et al (1997) Introducing dielectrophoresis as a new force field for field-flow fractionation. Biophys J 73(2):1118–1129View ArticleGoogle Scholar
- Huang Y et al (1999) The removal of human breast cancer cells from hematopoietic CD34+ stem cells by dielectrophoretic field-flow-fractionation. J Hematother Stem Cell Res 8(5):481–490View ArticleGoogle Scholar
- Peng H et al (2006) Dielectrophoresis field flow fractionation of single-walled carbon nanotubes. J Am Chem Soc 128(26):8396–8397View ArticleGoogle Scholar
- Park J et al (2005) An efficient cell separation system using 3D-asymmetric microelectrodes. Lab Chip 5(11):1264–1270View ArticleGoogle Scholar
- An J et al (2009) Separation of malignant human breast cancer epithelial cells from healthy epithelial cells using an advanced dielectrophoresis-activated cell sorter (DACS). Anal Bioanal Chem 394(3):801–809View ArticleGoogle Scholar
- Lee D et al (2013) A negative dielectrophoresis and gravity-driven flow-based high-throughput and high-efficiency cell-sorting system. J Lab Autom: 2211068213498385
- Nicoletti I et al (1991) A rapid and simple method for measuring thymocyte apoptosis by propidium iodide staining and flow cytometry. J Immunol Meth 139(2):271–279View ArticleGoogle Scholar
- Applegate R, Robert, et al (2004) Optical trapping, manipulation, and sorting of cells and colloids in microfluidic systems with diode laser bars. Opt Expr 12(19):4390–4398View ArticleGoogle Scholar
- Jr Applegate, Robert W et al (2006) Microfluidic sorting system based on optical waveguide integration and diode laser bar trapping. Lab Chip 6(3):422–426View ArticleGoogle Scholar
- Miltenyi S et al (1990) High gradient magnetic cell separation with MACS. Cytometry 11(2):231–238View ArticleGoogle Scholar
- He Z et al (2012) Isolation of human male germ-line stem cells using enzymatic digestion and magnetic-activated cell sorting Germline Development. Springer, New York, pp 45–57Google Scholar
- Seidl J, Knuechel R, Kunz-Schughart LA (1999) Evaluation of membrane physiology following fluorescence activated or magnetic cell separation. Cytometry 36(2):102–111View ArticleGoogle Scholar
- Hywel M, Green Nicolas G (2003) AC electrokinetics: colloids and nanoparticles. No. 2. Research Studies Press
- Hu X et al (2005) Marker-specific sorting of rare cells using dielectrophoresis. Proc Natl Acad Sci USA 102(44):15757–15761View ArticleGoogle Scholar
- Shim S et al (2013) Dielectrophoresis has broad applicability to marker-free isolation of tumor cells from blood by microfluidic systems. Biomicrofluidics 7(1):011808View ArticleGoogle Scholar
- Gascoyne PR et al (2013) Correlations between the dielectric properties and exterior morphology of cells revealed by dielectrophoretic field-flow fractionation. Electrophoresis 34(7):1042–1050View ArticleGoogle Scholar
- Lapizco-Encinas BH et al (2005) An insulator-based (electrodeless) dielectrophoretic concentrator for microbes in water. J Microbiol Meth 62(3):317–326View ArticleGoogle Scholar
- Gupta V et al (2012) ApoStream™, a new dielectrophoretic device for antibody independent isolation and recovery of viable cancer cells from blood. Biomicrofluidics 6(2):024133View ArticleGoogle Scholar
- Gascoyne Peter, Satayavivad Jutamaad, Ruchirawat Mathuros (2004) Microfluidic approaches to malaria detection. Acta Trop 89(3):357–369View ArticleGoogle Scholar
- Demierre N et al (2007) Characterization and optimization of liquid electrodes for lateral dielectrophoresis. Lab Chip 7(3):355–365View ArticleGoogle Scholar
- Cheng, I-Fang, Cheng-Che Chung, and Hsien-Chang Chang. “High-throughput electrokinetic bioparticle focusing based on a travelling-wave dielectrophoretic field.” Microfluidics and nanofluidics 10.3 (2011): 649-660
- Wang XB et al (2000) Cell separation by dielectrophoretic field-flow-fractionation. Anal Chem 72(4):832–839
- Lewpiriyawong Nuttawut, Yang C, Lam YC (2010) “Continuous sorting and separation of microparticles by size using AC dielectrophoresis in a PDMS microfluidic device with 3-D conducting PDMS composite electrodes.”. Electrophoresis 31(15):2622–2631
- Piacentini N et al (2011) Separation of platelets from other blood cells in continuous-flow by dielectrophoresis field-flow-fractionation. Biomicrofluidics 5(3):034122MathSciNetView ArticleGoogle Scholar
- Varadhachary G et al (2013) ApoStream™, a new dielectrophoretic device for antibody-independent isolation and recovery of circulating tumor cells from blood of patients with metastatic pancreatic adenocarcinoma. Cancer Res 73(8):1449View ArticleGoogle Scholar
- Kim U et al (2008) Multitarget dielectrophoresis activated cell sorter. Anal Chem 80(22):8656–8661View ArticleGoogle Scholar
- Kim Y et al (2011) Cantilever-type electrode array-based high-throughput microparticle sorting platform driven by gravitation and negative dielectrophoretic force. J Micromech Microeng 21(1):015015View ArticleGoogle Scholar
- Lee D et al (2016) Negative dielectrophoretic force based cell sorter with simplified structure for high reliability. Int J Precis Eng Manuf 17(2):1–5Google Scholar
- Wolff A et al (2003) Integrating advanced functionality in a microfabricated high-throughput fluorescent-activated cell sorter. Lab Chip 3(1):22–27View ArticleGoogle Scholar
- Ferrari Belinda C, Oregaard G, Sørensen SJ (2004) Recovery of GFP-labeled bacteria for culturing and molecular analysis after cell sorting using a benchtop flow cytometer. Microb Ecol 48(2):239–245View ArticleGoogle Scholar
- Soeth E et al (1996) The detection of disseminated tumor cells in bone marrow from colorectal-cancer patients by a cytokeratin-20-specific nested reverse-transcriptase-polymerase-chain reaction is related to the stage of disease. Int J Cancer 69(4):278–282View ArticleGoogle Scholar
- Büsch J et al (1994) Enrichment of fetal cells from maternal blood by high gradient magnetic cell sorting (double MACS) for PCR-based genetic analysis. Prenat Diagn 14(12):1129–1140View ArticleGoogle Scholar
- Schmitz B et al (1994) Magnetic activated cell sorting (MACS)—a new immunomagnetic method for megakaryocytic cell isolation: comparison of different separation techniques. Eur J Haematol 52(5):267–275MathSciNetView ArticleGoogle Scholar
- Adams Jonathan D, Kim U, Tom Soh H (2008) Multitarget magnetic activated cell sorter. Proc Natl Acad Sci 105(47):18165–18170View ArticleGoogle Scholar